El ClariVUE™10 es un sensor de turbidez (turbidímetro) de dispersión lateral sumergible que cumple con la norma ISO 7027. Comunica digitalmente por SDI-12, y es compatible con todos los dataloggers modernos de Campbell Scientific. La medida de la turbidez por dispersión lateral se puede utilizar como equivalente de la concentración de sedimentos en suspensión en ecosistemas acuáticos.Leer más
The ClariVUE™10 is an ISO 7027 compliant, side-scatter turbidity sensor. It returns data via SDI-12 to a data logger. Turbidity is a common surrogate measurement for suspended sediment concentration in aquatic systems. The ClariVUE™10 is designed to operate in fresh and saline environments. It will require regular maintenance to keep the optical window clean in high bio-fouling environments. The face of the sensor is made from copper to slow biological growth over the optical windows.
|Standards||Meets ISO 7027-1:2016 (side scatter) standard for turbidity measurements.|
|Compliance||SDI-12 v1.4 compliant|
|Operating Temperature Range||-2° to +40°C|
|Measurement Range||0 to 4000 FNU|
|Accuracy||±2% or 0.5 FNU (whichever is greater)|
|Power Requirements||9.6 to 18 Vdc|
|Measurement Time||9 s (with an M! instruction)|
|Housing Material||Delrin® plastic|
|Head Material||Marine-grade bronze|
|Connector Descriptions||Marine-grade bronze, gold-plated pins|
|Connector Description||Bronze 3-pin wet-mate|
|Cable Type||3-conductor, polyurethane jacket with water block|
|Optical Assembly||Sealed with double O-rings|
|Maximum Submersion Depth||30.48 m (100 ft) for the plastic|
|Diameter||30.1 mm (1.185 in.)|
|Length||166 mm (6.54 in.)|
|Weight||158.76 g (0.35 lb)|
|Quiescent||< 300 µA|
|Measurement||< 35 mA|
This example CR1000X program uses the M1! SDI-12 command to retrieve 100 burst measurements from a ClariVUE 5 or ClariVUE 10. The sensor makes 100 measurements to calculate turbidity statistics (median, mean, standard deviation, minimum, maximum) and mean temperature.
Número de FAQs relacionadas con ClariVUE10: 2
The most significant difference between backscatter and side-scatter measurements is the sample volume. A backscatter sensor has a much larger sample volume that helps measure higher turbidity values. The smaller sample volume of side-scatter sensors works better at lower turbidities.
It used to be that NTUs were the unit for all turbidity sensors. In 2007, an ASTM committee formed and established different units for each category of turbidity sensor technology, and there are many. Backscatter sensors such as the OBS-3+ and the OBS501 have units of FBU (Formazin Backscatter Units), whereas ISO 7027 side-scatter sensors have units of FNU (Formazin Nephelometric Units). The side-scatter OBS501 and the ClariiVUE™ sensors are part of this category. While sensors of different technical designs can be calibrated and measure Formazin the same, they will not make the same measurement in natural waters where size, shape, color, reflectivity, etc., of the particles vary. The ClariiVUE™ sensor measurements will not match the OBS-3+ measurements in natural waters.